Passive samplers provide an excellent opportunity to perform indicative measurements or establish a dense network of measuring sites. A drawback compared with conventional active measuring methods is the larger spread of results. This variation can, to a large extent, be attributed to the influence of temperature, sampler geometry and wind on sampling results. A proper design of sampler geometry and optimum choice of draught shield can reduce the influence of wind velocity on a badge type sampler to less than 10%. Wire mesh screens prove to be inadequate in damping turbulence. Filters give good results. Attention should be paid to the size and isolation value of the walls of the sampler to prevent thermal updrafts occurring within the sampler. Tube type samplers are less influenced by wind, provided that turbulence is prevented from influencing diffusion within the sampler.